A common question relates to the average life expectancy of the PLA002s permanent link adaptors. In short there is no definitive pre-determined life cycle of a cable although Fluke indicates around 5,000 insertions. However at www.testerhire.co.uk we have observed cables with good use to over 10,000 but on the flip side we have seen brand new cables damaged within just a few hundred. The main area for concern is the RJ45 tip with the eight gold coated contacts as these take the brunt of the insertion forces so are prone to damage and more often just wear. What happens is either the gold contacts became heavily tarnished and removes the highly conductive gold plating making contact with the RJ45 socket less than ideal and or the raised plastic profiles that separate each of the eight gold contacts becomes worn and damaged leading to cases where the plastic can "lean" to one side which makes good mating contact almost impossible. This usually happens to either pin1 or pair 1 or pin 8 or pair 4.
The Black sprung latch on top of the RJ45 housing also takes it's fair share of wear and often "rounds" the top edge and loses its firm spring action resulting in a less the positive seating in the RJ45 socket and often the "click" you use to hear when mated has no all but gone.
You can see the current autotest count of each PLA002s and channel adaptor by entering the special functions sections > Version. This will indicate the insertions of the cable or adaptor however a not of caution in that once you replace the DTX-PRP tip the autotest count will not reset so you need to keep a note of the autotest count when you changed the tip.
Another common source of failure is the retaining strain relief boots at both ends tend to became damaged and break after a period of time leading to strain being put on the pairs within either end causing a solder joint failure or such like which is almost impossible to detect or indeed repair. We have tried to eliminate this issue by strengthening each end with heatstrink and find this works well.
In summary keep an eye on your autotest count, visually inspect your RJ45 tips and strain relif boots at regular intervals and make sure you perform the self test and set reference at least once a day, all of which will help to reduce down time. We would also suggest the owners invest in spare DTX-PRP sets ready for when the need arises and our www.testerhire.co.uk customers can be sure we always have a good stock should the need arise.